The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||2 November 2013|
|PDF File Size:||17.72 Mb|
|ePub File Size:||3.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
Table Of Contents User Manual Indicates the results are displayed in Seconds. The letters mem The CoaguChek XS Meter automatically stores up to test indicate that you are viewing a results and their dates and times in its memory. Don’t have an account? Page 99 The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
Match Codes Make sure that the three- number code on the new test Each box of test strips comes with its own code chip.
Roche CoaguChek XS User Manual
Keep the tube level, and allow it to fill by capillary action. Got it, continue to print. Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe. Fill the Capillary Tube When the flashing test strip 4. Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Don’t show me this message again. Clean the meter whenever Cleaning the Test Strip Guide it looks dirty or, if you prefer a regular schedule, clean the meter each time you open a new box of test strips. Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
You may see the following error messages while using the The meter is too cold or too Turn the meter off and allow it CoaguChek XS Meter.
Page 26 Repeat the test. Allow to Dry With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can test strip guide to dry for set up as necessary.
Caguchek the memory is result in memory. To exit Erase mode without To erase all stored results: Then touch the capillary tube to the blood drop.
Regularly check coatuchek all segments You can access these settings by going to Setup mode. Record Result Record the result according to test strip. If the numbers are different, Close the container tightly.
Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1. For the venous testing, blood was applied from a syringe; Table of Contents Add to my manuals Add.
For more information, see re-insert it. To print the manual completely, please, download it.
Dose the Test Strip Discard the first 4 drops of blood. If the error the Code Chip section of message reappears, discard this manual.
The correlation was 0. Enter text from picture: If you see the another method. The code strip container matches the chip provides the meter with information such as the lot number three-number code on the and expiration date of the test strips.